Author Archives: Brian
I’ve been working on a lot of things of late, but none are yet in a state to show here yet. However, when talking to a colleague last week, I found it hard to explain why I do what I do: why I am so focused on metrology. But maybe I can explain it a […]
Last week, I talked about how to determine whether a flatfield correction was necessary. Data from a Bruker Hi-Star detector was shown to have very large local sensitivity deviations in the 2D detector image (of +/- 50%), and would therefore need a flatfield correction. So how to get one?
See the previous posts in this series here: Part 1, Part 2 and Part 3 Very recently, Bruker upgraded their SAXS instrument in our building. This is a typical 3-pinhole 2D SAXS system that was largely designed by J. S. Pedersen in Aarhus a decade or so ago. This particular version was at NIMS in […]
As indicated last week, some spare time has been spent trying to re-derive a correction for separating sample container scattering from the sample itself. Normally, a simple background subtraction suffices, but for those who want to go the extra mile (and in particular for those working with strongly absorbing sample containers and samples), you need […]
For today’s post, I had a few things planned, but haven’t been able to work anything out yet. So I’ll talk a little about these bits and pieces: Upcoming conferences and meetings, some future instrumentation plans, and a little about the Fancy Background Subtraction
Before I get started, let me just highlight the new small-angle scattering issue of J. Appl. Cryst. which has come out a few days ago. My contribution was unfortunately not accepted (and I haven’t worked much on it since due to rejection depression), but the journal is still very much worth a read! Now for […]
I feel I need to make a public apology: it turns out that I have been inaccurate in some of my recent papers on the explanation of the estimation of the uncertainties on the intensities (the uncertainties themselves, however, are correct, so I calculated it right but described it wrong). Here’s what happened:
A great many of the data correction issues in small-angle scattering can be resolved by trading a kidney for a PILATUS detector  (or similar ). These are detectors using a single crystal sensor as detection surface, and have extremely low noise, low distortion, high dynamic range and single photon sensing capabilities. However, as investigated […]
Over the last year or so, I have spent quite some words on the Monte Carlo method for extracting (particle) size distributions from scattering data. However, it may be more illustrative to show a short video of such a fit in action, simultaneously demonstrating the new GUI.
More about this next week, but Ingo has kindly provided a Windows executable version of one of the latest development builds of the Monte Carlo code (with GUI). While I have not yet had the chance to test it (being on a mac myself), please go ahead and get it from here! Some of you may […]